3281 peak hold function
QThe 3281 and 3282 also have a peak function. Is it possible to capture the crest value for a rush current as with the 3284 and 3285?
AAlthough the 3281 and 3282 offer peak hold capability through the combination of peak measurement and recording measurement, these instruments are not able to capture crest values 100% of the time. The 3281 updates the data every 250 ms. Measurement takes 128 ms, but calculation and display processing take up the remaining time, raising the possibility that a peak might be missed.