Current range of the 9694 when used with power meters or power quality analyzers (3169, 3196, 3197, PQ3100, PW3198, PW3360, PW3365)
Qs it possible to set the 50A range on the PW3360 when using the Clamp On Sensor 9694 (5A AC)?
The rated current of the 9694 is 5A AC, but its maximum input current is 50A continuous.
Therefore, the 9694 can be used in the 50A range with the PW3360.
However, please note that the accuracy of the 9694 is specified only up to 5A, and there is no defined accuracy for the 10A and 50A ranges.