POWER ANALYZER PW3390
V2.02 Aug 06, 2021
- If you are unable to download software, please check the following:
- Please register a my HIOKI account and then use it to log in to the my HIOKI service.
- Product firmware (product upgrades) and computer application software can only be downloaded for products you have registered on the my HIOKI service.
|Procedure on how to update the firmware (PW3390FirmwareUpdateProcedures.pdf)|
Version upgrade should be conducted by a qualified technician familiar with the functions and operations.
If you are uncertain as to how to proceed, please contact your local HIOKI distributor for assistance.
・All software provided here is freeware and the best version currently available.
・If the problems or errors during this download and install lead to malfunction, please contact your dealer or HIOKI representative.
・The ownership, intellectual right and all other associated rights relating to the software provided here belong to HIOKI E.E. Corporation.
If you have read and understand the information above and wish to proceed then please download the designated files.
As for version upgrade procedure, please download "PW3390FirmwareUpdateProcedures" which describes the preparation and procedure for upgrading firmware after downloading the files.
Precaution when updating software:
・To conduct the version upgrade, USB memory stick or HIOKI genuine CF card is required.
・During the version upgrade process, don’t turn off the power to the PW3390 or remove the CF card(USB memory tick) from the slot. Turning power off will result in failure to upgrade the PW3390 and the instrument must be returned to the HIOKI factory for repair.
Firmware for the model PW3390
The current newest software version is shown below.
Before you proceed
When selecting "SYSTEM" screen, the currently running firmware versionnumber will appear.
Version Upgrade History
V2.02 (August 2021)
1. Fixed bugs
-1. Fixed a bug that a response is returned when a ping command is sent to the old IP address after changing the IP address.
V2.01 (October 2019)
1. Fixed bugs
-1. Fixed a bug where the zero adjustment would not end when performing a zero adjustment while displaying certain screens.
-1. Changed the calculation method of the integrated power SUM value in multi-phase wirings in order to match to integration in buying and selling power. Power analyzer splits WP to WP+ and WP- according to the sign of the instantaneous SUM value.
-2. Other minor improvements were made.
V2.00 (May 2018)
1. Added functions
-1. Added a trend screen.
-2. Added a load function for screen shots.
-3. Added a FFT spectrum save function for noise measurements.
-4. Improved the following performance attributes for the noise analysis function:
Increased the maximum noise analysis frequency to 200kHz.
Decreased the minimum noise level from the current specifications by 20dB.
Improved calculation precision.
-5. Added the ability to manually adjust the motor phase zero adjustment value.
-1. When automatic save is turned OFF and waveforms are being saved while integration is being performed at an interval of 5s or shorter, missing points in the integration was found to occur. Also, regardless of the state of automatic save, if there are attempts to download waveforms using the PC application while integration is being performed at an interval of 5s or shorter, missing points across the entire measurement data file was found. This phenomenon has been fixed.
-2. Other minor bugs fixed.
-1. Replaced the text [deg.] with the symbol [°].
-2. Added support to events of synchronized measurements for the :HOLD and *TRG commands for the hold function via PC communication.
-3. Changed factory default settings for noise analysis related settings.
V1.01 (June 2017)
Improvements for production and inspection in the factory.
V1.00 (March 2017)