Evaluate the Amplification Quality of Sensors
Check the operation of amplifiers that receive the output of sensors such as a thermocouple and photodiode.
- Use the SS7012 DC Signal Source as a simulated signal source for sensors such as a thermocouple and photodiode and at the same time as a measuring instrument to measure the output voltage (current). Since generation and measurement are isolated, there is no interference between input and output.
- Use the SS7012 DC Signal Source as a source of signals of up to ±25 V and ±25 mA and at the same time measure the output of up to ±28 V and ±28 mA. The minimum resolution of generation and measurement is both 100 μV and 1 μA.
- The SS7012 is powered by batteries (Ni-MH or alkaline) or an AC adapter.
Add an output of the SS7012 DC Signal Source to the sensor amplifier circuit and confirm the return of the designed output.
You can download the PC sample software from the website.