CLAMP ON AC/DC HiTESTER 3287
Compact & Easy, One-Touch Maintenance on All Types of AC/DC Equipment
Hioki clamp meters are best-in-class measuring devices incorporating Hioki’s expertise in current sensing technology. The Hioki 3287 is an AC/DC current and voltage clamp meter that also measures continuity and resistance, featuring a double jaw lever and slim profile for easy access to crowded wires.
CAT III 600 V (Current)
CAT III 300 V (Voltage)
CAT II 600 V (Voltage)
- Accurately measure even small currents with 10 A range
- Voltage, resistance, and continuity check functions
Model No. (Order Code)
Compact & Easy
The 3287 performs accurate measurements of even small currents in the AC/DC 10.00 A range. In AC mode it also guarantees accuracy of true RMS measurements for frequencies from 10 Hz to 1 kHz. Handles low frequency to distorted currents including higher harmonics, simplifying maintenance of energy-saving equipment up to a maximum of 100 A.
|DC Current range||10.00/ 100.0 A, Basic accuracy: ±1.5 % rdg. ±5 dgt.|
|AC Current range||10.00/ 100.0 A (10 Hz to 1 kHz, True RMS)
Basic accuracy: ±1.5 % rdg. ±5 dgt.
|DC Voltage range||419.9 mV to 600 V, 5 ranges, Basic accuracy: ±1.3 % rdg. ±4 dgt.|
|AC Voltage range||4.199 V to 600 V, 4 ranges (30 to 500 Hz, True RMS)
Basic accuracy: ±2.3 % rdg. ±8 dgt.
|Resistance range||419.9 Ω to 41.99 MΩ, 6 ranges, Basic accuracy: ±2 % rdg. ±4 dgt|
|Crest factor||2.5 or less (150 A, 1000 V max.)|
|Other functions||Continuity: (50 Ω ±40 Ω) or less buzzer sounds, Data hold, Auto power save, Auto zero (DC A)|
|Display||LCD, max. 4199 dgt., Display refresh rate: 2.5 times/s|
|Power supply||Coin type lithium battery (CR2032) ×1, Continuous use 25 hours|
|Core jaw dia.||φ 35 mm (1.38 in)|
|Dimensions and mass||57 mm (2.24 in) W × 180 mm (7.09 in) H × 16 mm (0.63 in) D, 170 g (6.0 oz)|
|Included accessories||Coin type lithium battery (CR2032) × 1, Carrying case 9398 ×1, Test lead L9208 ×1, Instruction manual ×1|
Included accessories (2)
70 cm (2.30 ft) length
Secures one end of each test lead to the rear of the meter