LCR METER IM3536
If you are unable to download software, please check the following:
- Please register a myHioki account and then use it to log in to the myHioki service.
- Product firmware (product upgrades) and computer application software can only be downloaded for products you have registered on the myHioki service.
Version upgrade should be conducted by a qualified technician familiar with the functions and operations of model IM3536.
If you are uncertain as to how to proceed, please contact your local Hioki distributor for assistance.
・All software provided here is freeware and the best version currently available.
・If the problems or errors during this download and install lead to malfunction, please contact your dealer or Hioki representative.
・The ownership, intellectual right and all other associated rights relating to the software provided here belong to Hioki E.E. Corporation.
If you have read and understand the information above and wish to proceed then please download the designated files.
As for version upgrade procedure, please download ”Procedure on how to update the firmware” which describes the preparation and procedure for upgrading firmware after downloading the files.
Precaution when updating firmware:
・To conduct the version upgrade, USB memory stick is required.
・During the version upgrade process, don’t turn off the power to the IM3536 or don’t remove the USB memory stick from the slot. Turning power off will result in failure to upgrade the IM3536, and the instrument must be returned to the Hioki factory for repair.
Firmware for the Model IM3536
The current newest firmware versions are shown below. A version update is highly recommended if you are running older firmware.
Before you proceed
The version number of firmware can be checked by selecting the ”INFO” tab of the System screen.
Version Upgrade History
・In continuous mode, sending one of the following commands immediately after the instrument acknowledged the EOM signal caused an execution error.
:MEASure?, :MEMory?, :MEASure:CONTinuous:COMParator?
・During USB communications with a computer running Window 8, Window 8.1, or Windows 10, the instrument sent back two responses.
・Restarting the instrument after deleting a panel, which contains measurement condition data and correction data, may have disabled any more panels to be saved.
V1.03 (April 2016)
While setting to Internal trigger and measuring continuously, send a query command continuously by GP-IB I/F, measuring rarely stops.
V1.02 (January 2015)
1.When [ADJ]panel was loaded, the compensation could not be done correctly.
When a saved panel consisting of only compensation values was loaded, the panel seemed to be loaded on the screen; however, measured values were not compensated, if one of the following statuses of the loaded panel was different from one of those currently set in the instrument:
- Open compensation (OFF, SPOT including measurement frequency, or ALL)
- Short compensation (OFF, SPOT including measurement frequency, or ALL)
- Load compensation
2.Measurement status of ε or σ did not indicate ”DISP OUT”.
If the error message, ”DISP OUT” was displayed for a measured value of ε or σ on the screen, the response message for the measurement status query sent via communication was not ”3”, which represents that a measured value is out of the display range.
V1.01 (November 2014)