POWER ANALYZER PW6001
If you are unable to download software, please check the following:
- Please register a myHioki account and then use it to log in to the myHioki service.
- Product firmware (product upgrades) and computer application software can only be downloaded for products you have registered on the myHioki service.
Version upgrade should be conducted by a qualified technician familiar with the functions and operations of model PW6001.
If you are uncertain as to how to proceed, please contact your local Hioki distributor for assistance.
・All software provided here is freeware and the best version currently available.
・If the problems or errors during this download and install lead to malfunction, please contact your dealer or Hioki representative.
・The ownership, intellectual right and all other associated rights relating to the software provided here belong to Hioki E.E. Corporation.
If you have read and understand the information above and wish to proceed then please download the designated files.
As for version upgrade procedure, please download ”Procedure on how to update the firmware” which describes the preparation and procedure for upgrading firmware after downloading the files.
Precaution when updating software:
・To conduct the version upgrade, USB memory stick is required.
・During the version upgrade process, don’t turn off the power to the PW6001 or don’t remove the USB memory stick from the slot. Turning power off will result in failure to upgrade the PW6001, and the instrument must be returned to the Hioki factory for repair.
Software for the Model PW6001
The current newest software versions are shown below. A version update is highly recommended if you are running older software.
Firmware Version :V2.00
Before you proceed
Confirm the version number in the ‘CONFIG’ screen by pressing “SYSTEM” key.
Version Upgrade History
V2.00 (August 2016) Newest Version
-1. If the instrument received a command of a particular length via LAN, the communications failed subsequently.
-2. The operations in the following order with the synchronization control disabled caused the instrument to freeze: (1) configure the data update rate setting to 10 ms, (2) set to SLAVE MEAS (although the synchronization should have been disabled), and (3) display the COMMON setting screen.
-3. Changing the FFT settings of the slave instrument during the numerical synchronization caused an interruption of the integration in progress.
-4. If an AC/DC current sensor with a rated maximum current of 1 kA was connected to Probe 1, the integration mode could not be configured to DC.
-5. Other minor bugs fixed.
2. Added functions
-1. Added the D/A monitor graph screen.
-2. Added the X-Y plot screen.
-3. Added the FTP server function.
-4. Added the function that improves the measuring accuracy of phase difference in a high-frequency band.
-5. Communication via LR8410Link support
-6. Waveforms can now be saved in a binary format.
-7. Added the command that obtains measured values in the internal memory.
-8. Sensors with a rated maximum current of 5 A are now connectable.
-9. FFT analysis results can now be saved in a CSV format.
-1. It is now possible to configure the VT ratio and the CT ratio settings to 0.00001 at a minimum.
-2. It is now possible to load setting data that were created by using an instrument with an earlier version of the firmware installed and data including different sensor configuration.
-3. It is now possible to configure the sensor characteristics correction angle in increments of 0.01°.
-4. For the motor setting, the limitation that the pulse number shall be a multiple of half of the pole number is now removed.
-5. Reciprocal numbers of periods (frequency) between X and Y cursors are now additionally displayed when the cursors are used on the WAVE screen.
-6. Other enhancements.
V1.23 (January 2016)
-1. Changing zoom factor on the zoom screen by turning the rotary knob quickly before a displayed waveform reached the zoom area while waveform storage was in progress in real-time mode caused the instrument to freeze.
-2. Entering a value expressed as a decimal fraction preceded by two or more zeros, for example 000.951, 0035.852, and 00147.5, in the scaling (scale value) field for speed on the motor screen caused the alert “assert” to be displayed, resulting in the instrument freezing.
-3. The operations in the following order caused the alert “assert” to be displayed, resulting in the instrument freezing.
(1) Turn off the instrument, leaving it in any one of the 3-phase connection modes.
(2) Turn on the instrument.
(3) Change the connection mode to any one of the single-phase connection modes.
(4) Touch the U button on the Wave screen.
-4. Selecting a waveform by pressing the U or I button after selecting a channel of the slave on the Zoom screen by operating the channel keys during waveform synchronization caused the alert “assert” to be displayed, resulting in the instrument freezing.
-5. Performing integration while the auto-saving was activated with the screen displayed in English or Chinese, and then, performing an operation such as screen change after the elapse of a certain period of time caused the instrument to freeze.
-6.Other minor bugs fixed.
V1.22 (December 2015)
1. Fixed bugs
-1. In some cases while using Model PW6001-04 or PW6001-14, changing the connection mode to two circuits of 3P3W2M failed, resulting in an error.
V1.21 (October 2015)
1. Fixed bugs
-1. When the motor scaling value was set to a non-integer value, the motor waveform’s CSV output values were less than the actual values.
-2. When using the exponential mean and SLOW settings, there was a period of time during which invalid measured values were displayed immediately after the setting was changed.
-3. Saving a waveform while the sampling rate was set to 10 kHz caused the motor waveform’s sampling frequency to be displayed as 1 MHz.
-4. When averaging was enabled, in some cases the range was continually changed by the auto-range function, with the result that the range was not fixed (the measured value display showed “-------”).
-5. PW6001 ends up stopped by certain operations and communication command.
-6. Other minor bugs fixed.
2. Added functions
-1. User-defined calculation function (UDF)
-2. FFT calculation function
-3. Waveform cursor measurement function
-4. Waveform zoom function
-5. HTTP server function
-6. Screenshot display function
-7. Chinese language support
-1. The motor RPM scaling upper limit value has been changed from 9999.99 to 99999.9.
-2. The motor pole count upper limit value has been changed from 98 to 254.
-3. The file segment count upper limit used when outputting data has been changed from 20 to 100.
-4. Other enhancements.
V1.02 (March 2015)